1.Product consists of filter body and filter mesh. Filter mesh should have cells no more than 170 mkm. Filter bottom must have block stops.
- shortrun, spallings, crazing in filter body;
- shrinkage cavity, foreign inclusions, air bubbles in filter body;
- burr more than 0,2mm;
- foreign inclusions in filter mesh;
- full or partial tearing/displacement of filter mesh from filter body;
- oil spots, dirt in filter mesh and on the filter body;
blocking of holes for liquid outflow by the material
- Filter body – ABS (Acrylonitrile - butadiene - styrene) Medical grade.
- Filter mesh - PA (polyamide) Medical grade
Material must withstand sterilization by ethylene oxide and be compatible with human blood and blood components